The beginning with the name of the
Allah.
Who is kind and most merciful.

Slide 2:

Zaid Mahmood Farhat BEE-FA06-099
Department of Electrical Engineering

A.C.Bridges :

A.C.Bridges

Slide 4:

A.C.Bridges are those circuits which are used to measured the unknown resistances, capacitance and inductance.

Wheat Stone Bridge :

Wheat Stone Bridge Resistances can be measured by direct-current ,as shown in fig. a

Slide 6:

Fig. a
R 1 R 4 R 1 R 3 = R 2 R4
R 2 R 3

Slide 7:

Inductance and capacitance can also be measured by a similar four-arm bridge, as shown in fig.b.
In this case the alternating current source is employed by a vibration galvanometer.

Slide 8:

Fig.b

Maxwell’s inductance Bridge :

Maxwell’s inductance Bridge In the Maxwell’s inductance bridge ,there are two pure resistances used for balance relations but on other side or arms the two known impedances are used.
The known impedances and the resistances make the unknown impedances as Z1 and Z2.Such a network is known as Maxwell’s A.C.. Bridge. As shown in fig.

Slide 10:

Fig.
(R1 + jwL1 )R3 = (R4 + jwL4 )R2

Maxwell’s Wien Bridge :

Maxwell’s Wien Bridge In the Maxwell’s Wien Bridge the positive phase angle of the inductance may be compensated by the negative phase angle of the capacitance impedance put in the opposite arm.
The unknown inductance then becomes known in terms of the capacitance. As shown in fig.

Slide 12:

Fig.
R2 R4 + jwL1 R2 = R1 R3 + jwCR1 R2 R3

Anderson Bridge :

Anderson Bridge In the Anderson Bridge the unknown inductance is measured in terms of a known capacitance and resistance.
this method is capable of precise measurements of inductance over a wide range of values from a few micro-henrys to several henrys and is the best bridge method.

Hay’s Bridge :

Hay’s Bridge It is also a modification of the Maxwell’s Wien Bridge and is particularly useful if the phase angle of the inductive impedance is large.
In this case a comparatively smaller series resistance R1 is used instead of a parallel résistance.( which has to be of a very large value) as shown in fig.

Heavisible-Campbell Equal Ratio Bridge It is a mutual inductance bridge and is used for measuring self-inductance over a wide range in terms of mutual inductometer readings.
The connections for Heaviside’s bridge employing a standard variable mutual inductance.

Slide 18:

The primary of the mutual inductometer is inserted in the supply circuit and the secondary having self-inductance L2 and resistance R2 is put in arm 2 of the bridge.
The unknown inductive impedance having self-inductance of L1 and resistance R1 is placed in arm 1.
The other two arms have pure resistance of R3 and R4. as shown in fig.

Capacitance Bridge :

Capacitance Bridge We will consider only De Sauty bridge method of comparing two capacitances the bridge has maximum sensitivity when C2 = C3.
The simplicity of this method is offset by the impossibility of obtaining a perfect balance if both the capacitors are not free from the dielectric loss.
A perfect balance can only be obtained if air capacitors are used. as shown in fig.

Slide 21:

Fig.
C2= C3 R1
R2

Schering Bridge :

Schering Bridge Schering bridge used for the measurement of capacitance and dielectric loss of a capacitor.
It is a device for comparing an imperfect capacitor C2 in terms of a loss-free standard capacitor C1. As shown in fig.

Slide 23:

Fig.
C3 = C2 ( R1 / R2 ) R3 = R2 ( C1 / C2 )

Wien Series Bridge :

Wien Series Bridge It is a simple ratio bridge and is used for audio-frequency measurement of capacitance over a wide range. As shown in fig.

Slide 25:

Fig.
R1=R2R4/R3 C1=C4(R3/R2)

Wien Parallel Bridge :

Wien Parallel Bridge It is also a ratio bridge used mainly as the feedback network in the wide range audio-frequency R-C oscillators.
It is may be used for the measurement of the audio-frequency but it is not as accurate as the modern digital frequency meters. As shown in fig.

Slide 27:

Fig.
C2 = R2 = R3
C1 R1 R4

You do not have the permission to view this presentation. In order to view it, please
contact the author of the presentation.