logging in or signing up Digital Oscilloscope zaidmahmood Download Post to : URL : Related Presentations : Let's Connect Share Add to Flag Embed Email Send to Blogs and Networks Add to Channel Copy embed code: Embed: Flash iPad Dynamic Copy Does not support media & animations Automatically changes to Flash or non-Flash embed WordPress Embed Customize Embed URL: Copy Thumbnail: Copy The presentation is successfully added In Your Favorites. Views: 2332 Category: Science & Tech.. License: All Rights Reserved Like it (2) Dislike it (0) Added: June 09, 2009 This Presentation is Public Favorites: 2 Presentation Description No description available. Comments Posting comment... By: saajanpreet (57 month(s) ago) hey man can u pls send me this ppt to firstname.lastname@example.org i really require this for my project. it will help me a lot pls thank u Saving..... Post Reply Close Saving..... Edit Comment Close Premium member Presentation Transcript Digital Oscilloscope : Digital Oscilloscope UT2102C Zaid Mahmood Farhat : Zaid Mahmood Farhat BEE-FA06-099 Department of Electrical Engineering Digital Oscilloscope : Digital Oscilloscope Discussion Agenda : Discussion Agenda User’s Key points Vertical System Horizontal System Mathematics based Functions Trigger Control Addition Functional Bar Auto Setup Practical Example Scenario Front Panel and Basic Functionalities : Front Panel and Basic Functionalities Display Interface and Basic Functionalities : Display Interface and Basic Functionalities User’s Key points : User’s Key points We must want to know some important key points before using the equipment under consideration. General check Functional check Probe compensation Automatic settings for waveform display Getting to know the vertical system Getting to know the horizontal system Getting to know the trigger system General Check : General Check Check the unit for possible shipping damages Check the accessories Thorough inspection of the entire unit Functional Check : Functional Check Power on the unit and default setup Accessing signals Probe Compensation : Probe Compensation Adjust the probe on 1x or 10x according to the signals. If you see a “Undercompensation” or “Overcompensation” waveform display, adjust the variable capacitor on the probe with a screwdriver with non-metal handle, until a “Correct Compensation” waveform Automatic settings for waveform display : Automatic settings for waveform display Connect the signal to be measured to the signal input channel. Press [AUTO]. The oscilloscope will automatically set the vertical deflection factor, scanning time base and trigger mode. Vertical System : Vertical System Turn the vertical position knob to display the signal in the centre of the window. Change the vertical setups and check changes of status information. Horizontal System : Horizontal System Use the horizontal scale knob to change the horizontal time base setup and check any changes in status information. Range of horizontal scanning rate is 5ns~50s, in steps of 1-2-5. Use the horizontal position knob to adjust the horizontal position of the waveform window Press [MENU] to display the ZOOM menu. Shortcut Key Return to Zero! Trigger System : Trigger System Use the trigger level knob to change the trigger level. Open the [TRIGGER MENU] to change trigger setups. Press [F1] and select EDGE TRIGGER Press [F2] and set TRIGGER SOURCE to CH1 Press [F3] and set EDGE TYPE as RISING Press [F4] and set TRIGGER MODE as AUTO Press [F5] and set TRIGGER COUPLING as DC Press [50%] and set the trigger level at the vertical centre point of the trigger signal amplitude. Instrument Setups : Instrument Setups Vertical System : Vertical System Channel Coupling Channel Bandwidth Probe Rate Vertical VOLTS/DIV adjustment setup Waveform inversion setup Horizontal System : Horizontal System Horizontal Control Knob Main time base Window extension Mathematics based Functions : Mathematics based Functions Operating Math Functions FFT Window Reference Waveform Trigger Control : Trigger Control Edge Trigger Pulse Trigger Video Trigger Alternate Trigger Addition Functional Bar : Addition Functional Bar Acquisition mode? Sample peak detect Value [UTILITY] button to pop out the setup menu for alternative system functions. Auto Setup : Auto Setup Practical Example Scenario : Practical Example Scenario Automatic measurement of signal voltage and time parameters Your oscilloscope can automatically measure most display signals. To measure signal frequency and peak-to-peak value, follow the steps below: 1 .Press [MEASURE] to display the auto measurement menu. 2 .Press [F1] to enter the measurement type option menu. 3 .Press [F3] to select voltage. 4 .Press [F5] to go to page 2/4, then press [F3] to select measurement type: peak-to-peak value. 5 .Press [F2] to enter the measurement type option menu, then press [F4] to select time. 6 .Press [F2] to select measurement type: frequency. Questions? : Questions? Thank You! : Thank You! You do not have the permission to view this presentation. In order to view it, please contact the author of the presentation.