logging in or signing up CONDUCTOMETRY surreshkumarrr Download Post to : URL : Related Presentations : Share Add to Flag Embed Email Send to Blogs and Networks Add to Channel Uploaded from authorPOINT lite Insert YouTube videos in PowerPont slides with aS Desktop Copy embed code: Embed: Flash iPad Dynamic Copy Does not support media & animations Automatically changes to Flash or non-Flash embed WordPress Embed Customize Embed URL: Copy Thumbnail: Copy The presentation is successfully added In Your Favorites. Views: 1543 Category: Entertainment License: All Rights Reserved Like it (0) Dislike it (0) Added: February 06, 2013 This Presentation is Public Favorites: 0 Presentation Description No description available. Comments Posting comment... Premium member Presentation Transcript CONDUCTOMETRY : CONDUCTOMETRYCONDUCTOMETRY: CONDUCTOMETRY INTRODUCTION PRINCIPLE IMPORTANT DEFINITIONS & RELATIONS INSTRUMENTATIONS MEASUREMENT OF CONDUCTIVITY CONDUCTOMETRIC TITRATIONS ADVANTAGES & DISADVANTAGES OF CONDUCTOMETRIC TITRATIONS APPLICATIONS OF CONDUCTOMETRYPowerPoint Presentation: INTODUCTION: It is an electrochemical method of analysis concerned with electrical conductance through an electrolyte solution . ( or) It is defined has as determination or measurement of the electrical conductance of an electrolyte solution by means of a conductometer . electric conductivity of an electrolyte solution depends on : Type of ions ( cations , anions, singly or doubly charged Concentration of ions Temperature Mobility of ionsPowerPoint Presentation: PRINCIPLE: Based on the conductance of electrical current through electrolyte solutions similar to metallic conductors The electric conductance in accordance with ohms law which states that the stength of current( i )passing through conductor is directly porportional to potential difference & inversely to resistance. i =V/RPowerPoint Presentation: Important definitions & relations Conductance Specific conductance Molar conductance Equivalent conductance Resistance Specific resistancePowerPoint Presentation: Conductance:(G) ease with which current flows per unit area of conductor per unit potential applied & is reciprocal to resistance(R) G = I ⁄ R Specific conductance (K): conductance of the body of uniform length(l) &uniform area cross section(A) K = 1 ⁄ R × 1 ⁄ A Molar conductance: (^) Conductance of a solution containing 1 mole of the solute in 1000 cm of the solution which placed between two parallel electrodes which are 1 cm apart ^ = 1000 ⁄ CPowerPoint Presentation: Equivalent conductance: (^ eq ) specific conductance of the solution containing 1gm equivalent of solute in 1000cm 3 of solution. ^ eq =1000k ⁄ c eq Resistance (r): Is a measure of the conductors opposition to the flow of electric charge R =1/ G Specific resistance:( ρ ) Is resistance offered by a conductor of unit length and having unit cross section R ∞ l ⁄ APowerPoint Presentation: Instrumentation The instrument used for measurement of conductance are know as conductometers It consists of : Current source Alternating current source Conductivity cells Wide mouthed cells Cell for reactions producing precipitates Dip type cells ElectrodesPowerPoint Presentation: CURRENT SOURCE:- Mechanical high frequency AC generator by Washburn . Vreeland oscillator by Taylor and Acree. Vaccum tube oscillator by Hall & Adams. When electical potential is applied across electrodes two process occurs. Ions accumulate near the electrodes. Transfer of charge through the interface. Note : DC current is not employed in conductance measurement because Electrodes becomes polarised leading to high cell resistance.PowerPoint Presentation: Conductivity cells:- Made of pyrex or quartz and are fitted with two platinum electrodes. Should be placed in vessel containing water to maintain constant temperature Types : Wide mouthed cell Cell for reactions producing precipitation Dip type cellsPowerPoint Presentation: Wide mouthed cell:- Measurement of low conductance Wide mouthed fitted with an ebonite cover which has provisions for platinum electrodes and burettesPowerPoint Presentation: Cell for reactions producing ppts: Mainly used for ppt reactions Also wide mouthed fitted with ebonite cover which has provisions for burette ,electrode , stirrer Stirrer may be mechanical or magneticPowerPoint Presentation: Electrodes : Platinum sheets, each of 1 cm 2 are fixed at distance of 1 cm The surface is coated with platinum black to avoid polarization effects and increase effective surface area. Platinisation of electrodes is done by coating solution of 3% chlorplatinic acid and lead acetate on it to get uniform coating Electrodes usage depends on conductivity and concentration If conc is low then electrodes should be largely and closely packedPowerPoint Presentation: Measurement:- The instrument used to measure conductance is called conductance bridge or conductometer Classical circuit employed for measurement is wheatstone bridge All other work on this principle Various types are: Kohlrausch conductance bridge Direct reading conductance bridge Phillips conductance bridge Mullard’s conductance bridge Pye’s conductance bridgePowerPoint Presentation: Kohlrausch conductance bridge: Consists of a meter bridge XY with fixed resistors r’ & r” at both ends. One arm of bridge consists of resistance box ‘R’ & other arm with conductivity cell ’C’. Detector D is head phone while inductance coil ‘J’ is AC source which is operated by battery. Direct reading conductance bridge:- In this head phone is replaced by magic eye which is electronic devicePowerPoint Presentation: The set up for Measurement : consists of meter bridge LN attached to standard resistance R1 & unknown resistance R2 cell is connected to standard resistance to one side , meter bridge LN at other. The sliding contact with galvanometer (G) can be moved on the wire of meter bridge by means of jockey (M) so that resistance of unknown is balanced with that of standard. When galvanometer shows null deflection, the resistance of unknown is measured by following equation: ML ⁄ NL =R2 ⁄ R1 R2 =ML ⁄ NL × R1PowerPoint Presentation: Conductivity cell Solution Meter bridgePowerPoint Presentation: Hence conductivity of unknown solution: 1 ⁄ R2 =NL ⁄ ML × R1 The measured conductivity (1/R1) is not always equal to the specific conductivity of solution , because the physical configuration of platinum electrode i.e, length and area of electrodes varies from one another . Hence conductivity of solution is obtained by calculating a factor called “cell constant”.PowerPoint Presentation: Cell constant: Defined as ratio of distance between the two electrodes(l) to the area of electrodes(A) There fore, θ =1 ⁄ APowerPoint Presentation: CONDUCTOMETRIC TITRATIONS: INTRODUCTION : Is process of qualitative chemical analysis in which conc of sample is determined.Which is done by adding areagent( titrant ) of known conc in measured volumes to the sample (anylate )PowerPoint Presentation: TYRES OF CONDUCTOMETRIC TITRATIONS: Acid –base or neutral titrations Replacement or displacement titrations Redox titrations Precipitation titrations Complexometric titrations Non-aqueous titrationsPowerPoint Presentation: ACID- BASE OR NEUTRAL TITRATIONS: STRONG ACID - STRONG BASE EG: HCL vs NaOH STRONG ACID - WEAK BASE EG: HCL vs NH4OH WEAK ACID - STRONG BASE EG: CH3COOH vs NaOH WEAK ACID - WEAK BASE EG: CH3COOH vs NH4OHPowerPoint Presentation: Strong acid strong base: Fall in conductance due to replacement of high conductivity Hydrogen ions by poor conductivity sodium ions Rise in conductance due to increase in hydroxyl ionsPowerPoint Presentation: Strong acid- weak base: Fall in conductance due to replacement of hydrogen by ammonium ions Conductance remain constant due to supression of NH40H by NH4CLPowerPoint Presentation: Weak acid –Strong base: Initial decrease in conductance followed by increase due to NaOH Steep rise due to excess of NaOHPowerPoint Presentation: Weak acid- weak base: Increase in conductance due to excess of CH3COOH Constant conductance due to supression of NH4OH by CH3COOHPowerPoint Presentation: ADVANTAGE OF CONDUCTOMETRIC TITRATIONS: Does not require indicators since change in conductance is measured by conductometer Suitable for coloured solutions Since end point is determined by graphical means accurate results are obtained with minimum error Used for analysis of turbid suspensions, weak acids, weak bases, mix of weak & strong acidsPowerPoint Presentation: Dis advantages of conductometric titration: Increased level of salts in solution masks the conductivity changes , in such cases it does not give accurate results Application of conductometric titrations to redox systems is limited because, high concentrations of hydronium ions in the solution tends to mask the changes in conductancePowerPoint Presentation: Applications : Check water pollution in rivers and lakes Alkalinity of fresh water Salinity of sea water (oceanography) Deuterium ion concentration in water- deuterium mixture Food microbiology- for tracing micro organisms Tracing antibiotics Estimate ash content in sugar juices Purity of distilled and de - ionised water can determined Solubility of sparingly soluble salts like AgCl,BaSo4 can be detected Determination of atmospheric so 2, etimation of vanillin in vanilla flavourReferences: References Instrumental analysis by A. Skoog , F. James Holler and Stanly R. Crouch. Text book of pharmaceutical analysis , third edition by Dr.S.Ravi sankar . Presented by Suresh . G THE END : Presented by Suresh . G THE END You do not have the permission to view this presentation. In order to view it, please contact the author of the presentation.