CE2245-Surveying –II UNIT-I TACHEOMETRIC SURVEYING: CE2245-Surveying –II UNIT-I TACHEOMETRIC SURVEYING Prepared by C.MAKENDRAN Assistant Professor
PowerPoint Presentation: TACHEOMETRIC SURVEYING Tacheometry or tachemetry or telemetry is a branch of angular surveying in which the horizontal and Vertical distances of points are obtained by optical means as opposed to the ordinary slower process of measurements by tape or chain. The method is very rapid and convenient . It is best adapted in obstacles such as steep and broken ground, deep revines, stretches of water or swamp and so on, which make chaining difficult or impossible, The primary object of tacheometry is the preparation of contoured maps or plans requiring both the horizontal as well as Vertical control. Also, on surveys of higher accuracy, it provides a check on distances measured with the tape. Tacheometry (from Greek, quick measure), is a system of rapid surveying, by which the positions, both horizontal and vertical, of points on the earth surface relatively to one another are determined without using a chain or tape or a separate leveling instrument.
PowerPoint Presentation: Uses of Tacheometry The tacheometric methods of surveying are used with advantage over the direct methods of measurement of horizontal distances and differences in elevations. Some of the uses are: Preparation of topographic maps which require both elevations and horizontal distances. Survey work in difficult terrain where direct methods are inconvenient Detail filling Reconnaissance surveys for highways, railways, etc. Checking of already measured distances Hydrographic surveys and Establishing secondary control.
PowerPoint Presentation: INSTRUMENTS An ordinary transit theodolite fitted with a stadia diaphragm is generally used for tacheometric survey. - The stadia diaphragm essentially consists of one stadia hair above and the other an equal distance below the horizontal cross-hair, the stadia hairs being mounted in the ring and on the same vertical plane as the horizontal and vertical cross-hairs. Different forms of stadia diaphragm commonly used Stadia is a tacheometric form of distance measurement that relies on fixed angle intercept.
PowerPoint Presentation: The telescope used in stadia surveying are of three kinds: (1) The simple external-focusing telescope (2) the external-focusing anallactic telescope (Possor`s telescope) (3) the internal-focusing telescope. A tacheometer must essentially incorporate the following features: The multiplying constant should have a nominal value of 100 and the error contained in this value should not exceed 1 in 1000. The axial horizontal line should be exactly midway between the other two lines. The telescope should be truly anallactic. The telescope should be powerful having a magnification of 20 to 30 diameters. The aperture of the objective should be 35 to 45 mm in diameter to have a sufficiently bright image. For small distances (say upto 100 meters), ordinary levelling staff may be used. For greater distances a stadia rod may be used. A stadia rod is usually of one piece, having 3 – 5 meters length. A stadia rod graduated in 5 mm (i.e. 0.005 m) for smaller distances and while for longer distances, the rod may be graduated in 1 cm (i.e. 0.01 m).
PowerPoint Presentation: 10' 7"
PowerPoint Presentation: Different systems of Tacheometric Measurement: The various systems of tacheometric survey may be classified as follows: The stadia System (a) Fixed Hair method of Stadia method (b) Movable hair method, or Subtense method The tangential system Measurements by means of special instruments The principle common to all the systems is to calculate the horizontal distance between two points A and B and their distances in elevation, by observing The angle at the instrument at A subtended by a known short distance along a staff kept at B, and (ii) the vertical angle to B from A.
PowerPoint Presentation: (a) Fixed hair method In this method, the angle at the instrument at A subtended by a known short distance along a staff kept at B is made with the help of a stadia diaphragm having stadia wires at fixed or constant distance apart. The readings are on the staff corresponding to all the three wires taken. The staff intercept, i.e., the difference of the readings corresponding to top and bottom stadia wires will therefore depend on the distance of the staff from the instrument. When the staff intercept is more than the length of the staff, only half intercept is read. For inclined sight, readings may be taken by keeping the staff either vertical or normal to the line of sight. This is the most common method is tacheometry and the same ‘stadia method’ generally bears reference to this method.
PowerPoint Presentation: Subtense Method This method is similar to the fixed hair method except that the stadia interval is variable. Suitable arrangement is made to vary the distance between the stadia hair as to set them against the two targets on the staff kept at the point under observation. Thus, in this case, the staff intercept, i.e., the distance between the two targets is kept fixed while the stadia interval, i.e., the distance between the stadia hair is variable. As in the case of fixed hair method, inclined sights may also be taken. Tangential Method In this method, the stadia hairs are not used, the readings being taken against the horizontal cross-hair. To measure the staff intercept, two pointings of the instruments are, therefore, necessary. This necessitates measurement of vertical angles twice for one single observation.
PowerPoint Presentation: PRINCIPLE OF STADIA METHOD The stadia method is based on the principle that the ratio of the perpendicular to the base is constant in similar isosceles triangles. A O A 2 A 1 B 2 B 1 B C 2 C 1 C ) β In figure, let two rays OA and OB be equally inclined to central ray OC. Let A 2 B 2 , A 1 B 1 and AB be the staff intercepts. Evidently, OC 2 A 2 B 2 OC 1 A 1 B 1 OC AB = = = constant k = ½ cot β 2 This constant k entirely depends upon the magnitude of the angle β .
PowerPoint Presentation: In actual practice, observations may be made with either horizontal line of sight or with inclined line of sight. In the later case the staff may be kept either vertically or normal to the line of sight. First the distance-elevation formulae for the horizontal sights should be derived. Horizontal Sights: i . f 2 f 1 s O d M b C B A c a D Consider the figure, in which O is the optical centre of the objective of an external focusing telescope . Let A, C, and B = the points cut by the three lines of sight corresponding to three wires. b, c, and a = top, axial and bottom hairs of the diaphragm. ab = i = interval b/w the stadia hairs (stadia interval) AB = s = staff intercept; f = focal length of the objective
PowerPoint Presentation: f 1 = horizontal distance of the staff from the optical centre of the objective f 2 = horizontal distance of the cross-wires from O. d = distance of the vertical axis of the instrument from O. D = horizontal distance of the staff from the vertical axis of the instruments. M = centre of the instrument, corresponding to the vertical axis. Since the rays BOb and AOa pass through the optical centre, they are straight so that AOB and aOb are similar. Hence, f 1 s f 2 i = Again, since f 1 and f 2 are conjugate focal distances, we have from lens formula, 1 1 1 f f 2 f 1 + = Multiplying throughout by ff 1 , we get f 1 = f + f f 1 f 2 Substituting the values of in the above, we get f 1 s f 2 i = f 1 = f + f s i Horizontal distance between the axis and the staff is D = f 1 + d D = s + (f + d) = k . s + C f i
PowerPoint Presentation: Above equation is known as the distance equation . In order to get the horizontal distance, therefore, the staff intercept s is to be found by subtracting the staff readings corresponding to the top and bottom stadia hairs. The constant k = f/i is known as the multiplying constant or stadia interval factor and the constant (f + d) = C is known as the additive constant of the instrument. Determination of constant k and C The values of the multiplying constant k and the additive constant C can be computed by the following methods: 1 st method: In this method, the additive constant C = (f + d) is measured from the instrument while the multiplying constant k is computed from field observations: Focus the instrument to a distant object and measure along the telescope the distance between the objective and cross-hairs, The distance d between the instrument axis and the objective is variable in the case of external focusing telescope, being greater for short sights and smaller for long sights. It should, therefore be measured for average sight. Thus, the additive constant (f + d) is known. 1 1 1 f f 1 f 2 + =
PowerPoint Presentation: 3. To calculate the multiplying constant k, measure a known distance D 1 and take the intercept s 1 on the staff kept at that point, the line of sight being horizontal. Using the equation, D 1 = ks 1 + C or k = For average value, staff intercepts, s 2 , s 3 etc., can be measured corresponding to distance D 2 , D 3 etc., and mean value can be calculated. Note: In case of some external focusing instruments, the eye-piece-diaphragm unit moves during focusing. For such instruments d is constant and does not vary while focusing. D 1 – C s 2nd method: In this method, both the constants are determined by field observations as under: Measure a line, about 200m long, on fairly level ground and drive pegs at some interval, say 50 meters. Keep the staff on the pegs and observe the corresponding staff intercepts with horizontal sight. Knowing the values of D and s for different points, a number of simultaneous equations can be formed by substituting the values of D and s in equation D = k.s + C. The simultaneous solution of successive pairs will give the values of k and C, and the average of these can be found.
PowerPoint Presentation: For example, if s 1 is the staff intercept corresponding to distance D 1 and s 2 corresponding to D 2 we have, D 1 = k.s 1 + C . . . . . (i) and D 2 = k. s 2 + C . . . . . (ii) Subtracting (i) from (ii), we get k = D 2 – D 1 s 2 – s 1 . . . . . . . . . (1) Substituting the values of k in (i), we get C = D 1 - s 1 D 2 – D 1 s 2 – s 1 = D 1 s 2 – D 2 s 1 s 2 – s 1 . . . . . . . . . (2) Thus equation (1) and (2) give the values of k and C.
PowerPoint Presentation: Distance and Elevation formulae for Staff Vertical : Inclined Sight Let P = Instrument station; Q = Staff station M = position of instruments axis; O = Optical centre of the objective A, C, B = Points corresponding to the readings of the three hairs s = AB = Staff intercept; i = Stadia interval Ө = Inclination of the line of sight from the horizontal L = Length MC measured along the line of sight D = MQ’ = Horizontal distance between the instrument and the staff V = Vertical intercept at Q, between the line of sight and the horizontal line h = height of the instrument; r = central hair reading β = angle between the two extreme rays corresponding to stadia hairs.
PowerPoint Presentation: Draw a line A’CB’ normal to the line of sight OC. Angle AA`C = 90 0 + β /2, being the exterior angle of the ∆COA`. Similarly, from ∆COB`, angle OB`C = angle BB`C = 90 0 – β /2. h B A O D P A` C B` Q Q` M r V Ө β L Ө
PowerPoint Presentation: Since β /2 is very small (its value being equal to 17’ 11” for k = 100), angle AA’C and angle BB’C may be approximately taken equal to 90 0 . ∟ AA’C = ∟ BB’C = 90 0 From ∆ ACA’, A’C = AC cos Ө or A’B’ = AB cos Ө = s cos Ө ……….(a) Since the line A’B’ is perpendicular to the line of sight OC, equation D = k s + C is directly applicable. Hence, we have MC = L = k . A’B’ + C = k s cos Ө + C . . . . . . . (b) The horizontal distance D = L cos Ө = (k s cos Ө + C) cos Ө D = k s cos 2 Ө + C cos Ө . . . . . . (1) Similarly, V = L sin Ө = (k s cos Ө + C) sin Ө = k s cos Ө . sin Ө + C sin Ө V = k s + C sin Ө sin2 Ө 2 . . . . . . (2) Thus equations (1) and (2) are the distance and elevation formulae for inclined line of sight.
PowerPoint Presentation: (a) Elevation of the staff station for angle of elevation If the line of sight has an angle of elevation Ө , as shown in the figure, we have Elevation of staff station = Elevation of instrument station + h + V – r. (b) Elevation of the staff station for the angle of depression: Elevation of Q = Elevation of P + h – V - r
PowerPoint Presentation: Distance and Elevation formulae for Staff Normal : Inclined Sight h B A O D P C C` Q Q` M rcos Ө V Ө β L Ө L cos Ө rsin Ө Figure shows the case when the staff is held normal to the line of sight.
PowerPoint Presentation: Case (a): Line of Sight at an angle of elevation Ө Let AB = s = staff intercept; CQ = r = axial hair reading With the same notations as in the last case, we have MC = L = K s + C The horizontal distance between P and Q is given by D = MC’ + C’Q’ = L cos Ө + r sin Ө = (k s + C) cos Ө + r sin Ө . . . . . (3) Similarly, V = L sin Ө = (k s + C) sin Ө . . . . . (4)
PowerPoint Presentation: Case (a): Line of Sight at an angle of depression Ө rsin Ө M P h Ө A C C’ Q Q’ B C 1 D L cos Ө rcos Ө V O Figure shows the line of sight depressed downwards, MC = L = k s + C D = MQ’ = MC’ – Q’C’ = L cos Ө - r sin Ө D = (k s + C) cos Ө - r sin Ө . . . . . (5) V = L sin Ө = (k s + C) sin Ө . . . . . (6) Elevation of Q = Elevation of P + h – V – r cos Ө
PowerPoint Presentation: Thanking You