logging in or signing up Chest Tube Drainage liza_zpr Download Post to : URL : Related Presentations : Share Add to Flag Embed Email Send to Blogs and Networks Add to Channel Uploaded from authorPOINT lite Insert YouTube videos in PowerPont slides with aS Desktop Copy embed code: Embed: Flash iPad Dynamic Copy Does not support media & animations Automatically changes to Flash or non-Flash embed WordPress Embed Customize Embed URL: Copy Thumbnail: Copy The presentation is successfully added In Your Favorites. Views: 20167 Category: Education License: All Rights Reserved Like it (7) Dislike it (0) Added: February 03, 2009 This Presentation is Public Favorites: 6 Presentation Description NCM 102 Comments Posting comment... By: iamkei (31 month(s) ago) thank u! Saving..... Post Reply Close Saving..... Edit Comment Close By: eikrama (35 month(s) ago) Thank you so much Saving..... Post Reply Close Saving..... Edit Comment Close By: aaronfriel (38 month(s) ago) Thank you from Boise, Idaho!!!!! Saving..... Post Reply Close Saving..... Edit Comment Close By: xyrachan (41 month(s) ago) Big help to me. Thanks for this. :) Saving..... Post Reply Close Saving..... Edit Comment Close By: Drsawaung (43 month(s) ago) Very nice presentation . I'll refer it in my CME programme Dr. U Saw Aung Saving..... Post Reply Close Saving..... Edit Comment Close loading.... See all Premium member Presentation Transcript Concepts on Oxygenation : Concepts on Oxygenation Chest tubes & Water seal Drainage Indications for Chest tube drainage : Indications for Chest tube drainage To evacuate air or fluid from pleural space Pneumothorax > 20 % Hemothorax > 500cc Pneumohemothorax Pleural Effusion Fluid collections Mechanically ventilated patients with any size pneumothorax or hemothorax Indications for Chest tube Drainage : Indications for Chest tube Drainage Respiratory distress or ventilator dependence that does not permit thoracentesis Monitor ongoing air leak or blood loss Contraindications : Contraindications Coagulopathy or platelet dysfunction warrants caution Known or suspected mesothelioma Pure tuberculosis effusion Types & Purposes of Chest tubes : Types & Purposes of Chest tubes Pleural Tubes > the insertion of pleural tube permits removal of the air or bloody fluid & allows for reexpansion of the lung and restoration of the normal negative pressure in the pleural space . Types & Purposes of Chest Tubes cont’d. : Types & Purposes of Chest Tubes cont’d. 2. Mediastinal tubes > are placed to en - sure that fluid does not accumulate & put pressure on the heart but instead drains to facilitate healing Chest Tube Size : Chest Tube Size French sizing refers to circumference in millimeters Pure Pneumothorax : French 20 – 24 Fluid : French 28 tubes are recommended French 32 or larger tubes tend to cause too much discomfort unless a rib resection is performed Principles of Chest Drainage : Principles of Chest Drainage Principles Involved : Principles Involved Gravity positive pressure Suction Water – seal principle TYPES OF CHEST TUBE DRAINAGE SYSTEM : TYPES OF CHEST TUBE DRAINAGE SYSTEM Slide 12: SINGLE – BOTTLE WATER SEAL SYSTEM The end of the drainage tube from the patient’s chest is covered by a layer of water , which permits drainage of air & fluid from the pleural space , but does not allow air to move back into the chest . Functionally , drainage depends on gravity , on the mechanism of resopiration , & if desired , on suction by addition of controlled vacuum 2. TWO – BOTTLE WATER SEAL SYSTEM consists of the same water – seal chamber , plus a fluid – collection bottle Drainage is similar to that of a single unit , except that when pleural fluid drains , the underwater – seal system is not affected by the volume of the drainage Effective drainage depends on gravity or on the amount of suction added to the system Slide 13: 3. THREE- BOTTLE WATER SEAL SYSTEM There’s an addition of a 3rd bottle to control the amount of suction applied The amount of suction is determined by the depth to which the tip of the venting glass tube is submerged in the water Drainage depends on the gravity or the amount of suction applied , which is controlled by the manometer bottle The mechanical suction motor / wall suction creates & maintains a negative pressure throughout the entire closed drainage system 4. PLEUR – EVAC WATER SEAL SYSTEM Is a three- chambered system that utilizes the same basic principles as the classic three-bottle system You do not have the permission to view this presentation. In order to view it, please contact the author of the presentation.