logging in or signing up The Evolution of the Epidural Needles-4 aSGuest61907 Download Post to : URL : Related Presentations : Let's Connect Share Add to Flag Embed Email Send to Blogs and Networks Add to Channel Copy embed code: Embed: Flash iPad Dynamic Copy Does not support media & animations Automatically changes to Flash or non-Flash embed WordPress Embed Customize Embed URL: Copy Thumbnail: Copy The presentation is successfully added In Your Favorites. Views: 1514 Category: Science & Tech.. License: All Rights Reserved Like it (2) Dislike it (0) Added: August 21, 2010 This Presentation is Public Favorites: 0 Presentation Description No description available. Comments Posting comment... Premium member Presentation Transcript The Evolution of the Epidural Needles : The Evolution of the Epidural Needles Joseph Eldor, MD All Epidural Needles Are Not Created Equal : All Epidural Needles Are Not Created Equal Spectral analysis of epidural needles : Spectral analysis of epidural needles Slide 4: Epidural needles are composed of Iron (69%), Chromium (18%), Nickel (9%), Manganese (1.5%), Silicone(0.5%) and Molybdenum (between 0.2% to 2.7%). Different needles – different compositions… Hardness of the needles and stilletes : Hardness of the needles and stilletes Slide 6: Epidural needles differ also by the hardness of the needles and stilletes. Slide 7: For example, BD 18G Tuohy needle is 383 VPN Vs. BD 18G Crawford needle’s 507 VPN. Again, different needles – different composition… Dimensions and clearance of needles : Dimensions and clearance of needles Slide 9: Epidural needles differ also by their dimensions. For example, the inner diameter of the BD 18G Tuohy needle is 1.053 mm while the Portex 18G Tuohy needle’s is 1.220 mm… Slide 10: However, Epidural anesthesia is an integral part of today’s practice of Anesthesiology. Slide 11: A change in needle design was significant among the improvements. Slide 12: Epidural needles are often very broadly referred as “Tuohy needles” even though their needle tip configuration is much closer to that of a Hustead needle design. Slide 13: In 1901 the French Radiologist Jean Anthanase Sicard described injecting dilute solutions of cocaine through the sacral hiatus to treat patients suffering from severe intractable pain or lumbago. Dr. JEAN ANTHANASE SICARD (1872-1929) : Dr. JEAN ANTHANASE SICARD (1872-1929) Slide 15: Fernand Cathelin reported similar work 3 weeks later than Sicard. He recognized that sacral injections of cocaine might also be used for surgery. Slide 16: In 1921 a Spanish surgeon Fidel Pagés described a lumbar approach to epidural anesthesia. Slide 17: In 1933 an Italian surgeon Archile Mario Dogliotti performed abdominal surgery with single shot lumbar epidural anesthesia. Slide 18: Most medical historians date the regular use of epidural anesthesia from Dogliotti’s article: Dogliotti AM. A new method of block: segmental peridural spinal anesthesia. Am J Surg 1933;20: 107-18 Slide 19: However, a Romanian obstetrician ,Eugene Aburel, injected chinocaine through a silk ureteral catheter to block the lumboaortic plexus of laboring women two years before Dogliotti, in 1931. Slide 20: In the USA, Hingson and Edwards devised a method for continuous caudal anesthesia and used it on 33 laboring patients in 1942. Slide 21: Two years later, in 1944, Hingson and Southworth described a lumbar approach for continuous epidural anesthesia using a 15 gauge Barker spinal needle and a silk ureteral catheter “to but not into the peridural space”. They obtained satisfactory anesthesia in only 10 of 16 patients. BARKER NEEDLE : BARKER NEEDLE Slide 23: Ralph L. Huber designed in 1946 a needle with a directional tip which allowed anesthesiologists to direct the catheter as it exited the needle tip (US patent 2 409 979. October 22, 1946). Slide 24: Although Huber intended this needle for IV and tissue injections, Edward B. Tuohy recognized that the directional point might facilitate placement of spinal catheters. Slide 25: Tuohy also added a stylet, thereby hoping to further decrease the risk of skin plugging. Slide 26: The Cuban anesthesiologist, Manuel Martinez Curbelo, described in 1949 the use of a 16 gauge Huber (Tuohy) needle with a 3.5 F silk ureteral catheter for continuous segmental lumbar peridural anesthesia. Slide 27: Charles E. Flowers, an obstetrician at Johns Hopkins University, altered the Huber (Tuohy) needle by blunting the bevel and designing the stylet to protrude past the needle tip in 1949. Slide 28: However, Robert Hustead, another Hopkins colleague, called the Tuohy-Flowers modification a “technical nightmare”… Slide 30: However, Flower’s idea of a blunt tip did have appeal, and it became part of further enhancements by Crawford, Weiss, Sprotte and even Hustead himself. Slide 31: Robert Hustead made his own modifications to the Tuohy-Huber needle by hand by using a stone and a needle sharpener… Slide 32: Hustead sanded off the sharp tip of the original Tuohy-Huber needle and changed the angle of the bevel. The result was a needle opening that did not exceed 2.7 mm in length, with an angle of the needle bevel of 12º-15º. Robert F. Hustead (born: 1928) : Robert F. Hustead (born: 1928) Slide 34: Crawford described in 1951 the use of an epidural needle with a straight tip in 677 patients for thoracic surgery.The Crawford epidural needle was a Quincke type with an extremely short, and thus a very blunt, bevel. The bevel of his needle was very flat (60º if measured from the longitudinal axis of the needle). Oral Bascom Crawford (born: 1921) : Oral Bascom Crawford (born: 1921) CRAWFORD EPIDURAL NEEDLE : CRAWFORD EPIDURAL NEEDLE Slide 37: Jess Weiss dulled the needle tip and added “wings” to make it easier to grasp the needle with both hands for placement. Jess Bernard Weiss (born: 1917) : Jess Bernard Weiss (born: 1917) WEISS TUOHY NEEDLE : WEISS TUOHY NEEDLE Slide 41: Jürgen Sprotte of Würzburg, Germany developed in 1979 a pencil point needle for spinal anesthesia, similar to one designed by Hart and Whitacre but with a noticeable difference in its tip geometry: the original Whitacre needle had a short, cylindrical-tip configuration, whereas Sprotte used the olive-shaped, rounded-tip geometry. Slide 42: In 1987 Sprotte started to experiment to make his needle more suitable for epidural use. He added a special plastic wedge to the inside of the needle tip that would direct an epidural catheter toward the lateral needle hole. Jürgen Sprotte (born: 1945) : Jürgen Sprotte (born: 1945) Epidural Sprotte Spezial Needle : Epidural Sprotte Spezial Needle Other Epidural Needles : Other Epidural Needles Slide 46: Cheng needle was described in 1957. It was the first epidural needle with centimeter markings to indicate depth (US patent no. 2 922 420). Slide 48: Unfortunately, the Cheng needle was described as “cumbersome” to use and “extremely difficult to effect”. Slide 49: There are also the Wagner needle (1957), the Crawley needle (1968), the Foldes needle (1973) and the Bell needle (1975) – all variants of the Huber design with a blunted tip of varying sharpness. Slide 50: There is the Brace needle, a Crawford variant. Slide 51: There is the Lutz epidural needle (1963) with a pencil point design for single-shot epidural use. Lutz epidural needle : Lutz epidural needle Slide 53: There is the Scott needle (1985), a Tuohy needle with a Luer lock hub. Aldrete 25G Epidural Needle : Aldrete 25G Epidural Needle Pain Digest 8:260-263, 1998 Eldor, Coombs and Torrieri Needle : Eldor, Coombs and Torrieri Needle Eldor and Torrieri described in separate letters, in 1988, an epidural needle with a spinal needle attached to it. Through the spinal needle a longer spinal needle is inserted into the subarachnoid space, while an epidural catheter is introduced through the epidural needle into the epidural space. A few months before the publication of these letters, Coombs applied for a patent on the same device. Eldor Needle : Eldor Needle The Eldor needle was first described in 1990. The Eldor needle is a combined spinal-epidural needle which is composed of an 18 gauge epidural needle with a 20 gauge spinal conduit. This is a specialized needle for the combined spinal-epidural anesthesia. The spinal conduit of the Eldor needle serves as a break at the ligamentum flavum for an inadvertent insertion into the spinal space : The spinal conduit of the Eldor needle serves as a break at the ligamentum flavum for an inadvertent insertion into the spinal space Eldor CSEA needle : Eldor CSEA needle Needle-through-needle technique : Needle-through-needle technique Coates from England and Mumtaz, Daz and Kuz from Sweden, in two separate letters in the same issue of Anaesthesia, in 1982, first described the insertion of a long spinal needle through the epidural needle for performing the combined spinal-epidural anesthesia. Coates found the technique "simple, reliable and relatively quick to perform". After injecting the anesthetic solution into the subarachnoid space, the spinal needle is withdrawn and an epidural catheter is threaded through the same lumen of the epidural needle, through which the spinal needle was inserted. Metallic microparticles are produced by the friction between the two needles in the needle-through-needle technique : Metallic microparticles are produced by the friction between the two needles in the needle-through-needle technique Inadvertent epidural catheter protrusion (B) through the dural hole made by the spinal needle (A) in the needle-through-needle technique : Inadvertent epidural catheter protrusion (B) through the dural hole made by the spinal needle (A) in the needle-through-needle technique Inadvertent epidural needle protrusion (A) and catheter insertion (B) into the subarachnoid space in the needle-through-needle technique : Inadvertent epidural needle protrusion (A) and catheter insertion (B) into the subarachnoid space in the needle-through-needle technique A. Needle-through-needle technique: The Whitacre pencil point spinal needle deflects by 10 degrees from the tip of the epidural needle.B. Eldor needle technique: The double-hole pencil point spinal needle exits the spinal conduit of the Eldor needle in a straight direction and parallel to the epidural tip. : A. Needle-through-needle technique: The Whitacre pencil point spinal needle deflects by 10 degrees from the tip of the epidural needle.B. Eldor needle technique: The double-hole pencil point spinal needle exits the spinal conduit of the Eldor needle in a straight direction and parallel to the epidural tip. Huber needle technique : Huber needle technique Huber, the inventor of the "Tuohy" epidural needle, also patented in 1953 an hypodermic needle with an "auxiliary outlet being disposed in transverse alignment with the channel outlet". Hanaoka described in 1986 its use in 500 patients. This needle has a very small hole behind the epidural needle tip ("back eye"). A small gauge spinal needle is inserted through that hole and punctures the dura. After withdrawing the spinal needle an epidural catheter is introduced through the epidural needle. Espocan Tuohy Needle : Espocan Tuohy Needle The “Twin Theory” : The “Twin Theory” CSEA: "To paint the fence" from both its sides : CSEA: "To paint the fence" from both its sides Slide 72: The first technical advancement over the early years of epidural anesthesia was the downsizing of the catheter and needle combination used. Slide 73: Tuohy’s idea of using a lacquered silk catheter for continuous spinal anesthesia was stimulated by his neurosurgical colleague, Dr. Love, who already used this catheter for subarachnoid drainage at the Mayo Clinic. Slide 74: Whereas Love had advocated the use of a 5F ureteral catheter in combination with a 13 gauge Barker needle, Tuohy used a 15 gauge Barker needle and a 4F ureteral catheter in 1944. Edward Boyce Tuohy (1908-1959) : Edward Boyce Tuohy (1908-1959) Slide 76: Some years later, in 1949, Curbelo advocated the use of a 3.5F silk catheter with a 16 gauge Tuohy needle. Slide 77: The use of plastic catheters was first described by Flowers et al. in 1949. Slide 78: The first polymer (plastic) was polyethylene. It was soon replaced by polyvinyl chloride because of its low melting point, which, similar to the lacquered silk catheter, made it prone to swelling and deformity with sterilization. Slide 79: More recent polymers are nylon, Teflon, polyurethane and silicone. Slide 80: Most of the epidural catheters used in the USA are of the single terminal hole, while most of those used in the UK are of the three lateral holes with a blunted end. Slide 81: Curbelo in 1947 was the first to use an epidural catheter for epidural anesthesia. It was of the end hole type. Slide 82: Lee in 1962 described an epidural catheter which its "tip is non-patent and smooth, to facilitate insertion, solution enters the space from a small opening 1 cm from the tip". Slide 83: Skinner in 1966 described another epidural catheter with "the tip of the cannula...is blunt...and has a hole in its end. There is a second hole, 3 mm from the distal end". He said that "with the hole in this position, kinking is impossible with the stresses involved in normal use". Slide 84: Collier and Gatt described in 1993 an epidural catheter, with a closed end and three holes spaced at approximately 2, 3 and 4 mm from the smooth closed tip. They suggested that the close spacings of the holes will eliminate the complication of multi-compartment block (epidural, spinal or intravenous), as seen with earlier multi-hole epidural catheters. Slide 85: Michael et al. compared between the open-end (single hole) and closed-end (three lateral holes) epidural catheters. They found that the open-end catheters "caused an unacceptably high incidence of unsatisfactory sensory blockade". Slide 86: Collier and Gatt also found the incidence of unsatisfactory blocks with terminal hole catheter to be "unacceptably High" (32%) when compared with the three lateral holes catheter (12%). Slide 87: Eldor described in 1998 the CEMLH epidural catheter that has 7 holes within its 1.5 cm head: One at the tip while the 6 lateral holes are arranged circumferentially from each other (US patent no. 5 800 407). Slide 89: Evaluation of epidural needles Slide 90: Dural tissue trauma and cerebrospinal fluid leak after epidural needle puncture: effect of needle design, angle, and bevel orientation. Slide 91: Cerebrospinal fluid leak after puncture was influenced most by epidural needle gauge. Leak rate was significantly less for the 20-gauge Tuohy needle. Slide 92: Angle PJ, Kronberg JE, Thompson DE, Ackerley C, Szalai JP, Duffin J, Faure P. Dural tissue trauma and cerebrospinal fluid leak after epidural needle puncture: effect of needle design, angle, and bevel orientation. Anesthesiology 2003 Dec;99(6):1376-82. Slide 93: Electron microscopic analysis of particles from surgical gloves and their possible introduction into the epidural space during epidural anesthesia Slide 94: Particles adhering to gloves can be drawn into the epidural space during continuous epidural anesthesia. All unnecessary manipulation should therefore be avoided, and the portion of the catheter to be inserted into the epidural space should not be touched in order to prevent possible nonspecific meningeal inflammatory responses. Slide 95: Reina MA, Lopez Garcia A, Aguilar JL, Palacios Martin R. Electron microscopic analysis of particles from surgical gloves and their possible introduction into the epidural space during epidural anesthesia. Rev Esp Anestesiol Reanim. 1999 Feb;46(2):60-6 From the skin to the epidural space : From the skin to the epidural space Slide 97: The distribution of distance from the skin to the epidural space in obstetric patients (n=2,123) was: < 3 cm - 0.3%; 3 to < 4 cm - 15%; 4 to < 5 cm - 47.3%; 5 to < 6 cm - 28.6%; 6 to < 7 cm - 6.9%; 7 to < 8 cm - 1.4%; and > 8 cm - 0.5% Slide 98: The mean value of the distance from skin to the epidural space in term parturients is influenced by the patient`s position during epidural needle placement. It is 4.44 ± 0.82 cm in the sitting position and 5.03 ± 1.05 cm in the lateral position. Slide 99: Spinal dura mater thickness at the lumbar region is 0.5 mm. Slide 100: Using magnetic resonance imaging (MRI) of 39 patients Westbrook et al. found that the depth of the extradural space at the level of T12 is 0.83±0.195 cm. Westbrook JL, Renowden SA, Carrie LES. Study of the anatomy of the extradural region using magnetic resonance imaging. Br J Anaesth 1993;71:495-498 Slide 101: Cerebrospinal fluid is formed at the rate of 400 ml/day. Slide 102: The average adult has a total of 120 to 150 ml of fluid, of which only 20 to 35 ml are in the spinal portion of the subarachnoid space. In parturients, this volume in the spinal canal is decreased. Slide 103: Valsalva, in 1682, was the first to remark on the CSF when he cut open the spine of a dog and noticed the liquid "which in all its aspects resembles that which is found in the joints". Slide 104: Meme et al. using an epiduroscope could visualize the pulsation of the dura synchronized to heart rate. Meme E, Amici M, Ricci L, Merletti F. Epidural space: anatomical living view. International Monitor on Regional Anaesthesia 1993;96 Slide 105: The human dura mater shows its fibers neither longitudinally nor parallel directed. It has thin collagen fibers and thick elastic ones with wavy characteristics, grouped in ribbons or sheets. Human dura mater, Epidural surface (x3,500) : Human dura mater, Epidural surface (x3,500) CSEGA : CSEGA Each one of the three kinds of anesthesia (spinal, epidural, general) has its advantages and disadvantages. A new concept of combined spinal-epidural-general anesthesia (CSEGA) is illustrated with the objective of producing a new kind of anesthesia. Slide 108: CSEGA is a new concept in anesthesia. The mixing of regional anesthesia with general anesthesia affords the anesthesiologist the opportunity to lower the local anesthetic doses and avoid using many kinds of intravenous drugs. Slide 109: Zoric S, Stamenkovic D, Stevanovic S, Malenkovic V, Dikic SD, Randelovic T, Bilanovic D. Combined spinal epidural and general anesthesia in abdominal surgery. Med Arh. 2003;57(4 Suppl 1):21-8 Slide 110: "All evolution in thought and conduct must at first appear as heresy and misconduct“ George Bernard Shaw (1856-1950) For Example:… : For Example:… Slide 112: THANK YOU You do not have the permission to view this presentation. In order to view it, please contact the author of the presentation.